Bluetooth blood glucose meter
standard exporting packing or as clients request.
CE approved android bluetooth blood glucose meter
Blood Sample Type: Freshly Developed Microvascular Whole Blood For Blood Volume Range: 25% to 65%
Sample Volume Requirement: 1 μL Time Setting: You can set the time and date
Operating temperature: 10 ° C to 40 ° C (50 ° F to 104 ° F) Test scenario setting: Markable values for pre-meal and post-meal
Operating humidity: relative humidity 20% to 80% Power: 3V lithium battery one (battery type CR2032)
Measuring range: 1.1 ~ 33.3mmol / L
Battery life: about 1000 times the test (the actual battery life according to the situation and the battery brand is different)
Test time: 8 seconds hypoglycemia
warning function: the default blood glucose 3.9 ~ 17.7mmol / L, out of range that shows high, low blood sugar warning signs
Memory capacity: 180 pen memory power saving device: 3 minutes no action automatically shut down
1. on the blood glucose meter program and APP
A: blood glucose meter program and APP by our R & D department independent research and development completed, APP free of charge.
2. on the Bluetooth serial protocol
A: If your company needs to use their own APP software, our company can provide Bluetooth serial protocol.
3. on the Bluetooth module
A: Our mainframe chip has been embedded in RX, TX serial port transmission protocol, just add the Bluetooth serial protocol.
Such as your company has a research and development team to use their own development of the Bluetooth module, please buy the Bluetooth version of the blood glucose meter,
Check if your Bluetooth IC has a space on the circuit board.
4. on the test paper problem
A: The test paper is a box of 25 pieces of equipment, the box out of the test paper, more than five minutes test paper easy to fail.
Because the test paper contains biochemical technology, a long time in the air exposed vulnerable to air, temperature, humidity
, Resulting in partial oxidation of the reactive enzyme inside, thereby reducing the accuracy of the measurement.
5. on the machine
A: 1, the test paper in the machine, if E23, indicating the test paper damage or moisture, if E24, indicating that the test paper has been used, or the test steps are not correct.
2, after measuring blood sugar, less than 3.9mmol / L or higher than 17.7mmol / L will appear hypoglycemia Warning. Please refer to the manual.
6. on the Bluetooth power saving mode
A: Our design of the Bluetooth blood glucose meter, its Bluetooth low power transmission (BLE) power consumption of 2 ~ 3μA, not working state is dormant, working state was awakened, designed to meet environmental
7. on the measurement of blood glucose meter measurement standards
A: The blood glucose meter in the market to measure the blood glucose required to collect blood are capillary blood, how to measure whether the measurement accuracy? First analysis of blood sugar content is the number of blood need to analyze the blood inside Pulp, the general measure of the standard is 3 or more hospital blood glucose (plasma) analyzer out of the results for
8. on the blood glucose meter simulation machine (prototype)
A: Our Bluetooth blood glucose meter has been mass production, and accept the scheduled, but also that our product is the final version, such as your company to be tested, I provide the Secretary for testing, no blood, with our design